I am using Visual Studio Premium on Windows 7 bit. Legal Privacy Terms Trademark Acknowledgements. I would appreciate your help. There must be something that I am not seeing. Computer Software Current Version: I have Agilent A too and drivers of course. This program sets the A or A for 10 DCV measurements on the 10V range with a 1 second delay between the measurements.
|Date Added:||10 August 2012|
|File Size:||50.44 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I mean you do not need a lot of code to communicate with the device. There is no such thing as Agilent Keithley.
This program sets the A, A for 10 DCV measurements on the 10V range with a 1 second delay agilent 34410a usb the measurements. You can communicate agilent 34410a usb this codes i gave or you can communicate with the initialize function every device driver has one first and use other functions for example measurement functions This is my frist time writing a program to retrieve measurements from this device.
I would have thought it would have been obvious that the two are completely different. BenchVue software offers multiple instrument measurement visibility and data capture from a PC or mobile device with no agilent 34410a usb.
Message 3 of 6.
Message 2 of 6. Contains a complete collection of all BenchVue instrument control, automation and analysis apps.
Communicating with Agilent A via USB – Discussion Forums – National Instruments
Message 6 of 6. What commands should I be using? At first communicating with the device is really easy. Agilent 34410a usb am not sure what I am doing wrong. Easily control your DMM to quickly build automated tests, log measurements, and save valuable time.
Message 4 of 6.
34410A Digital Multimeter, 6½ Digit
Message 5 of 6. Agilsnt measurements are started with BUS trigger i. If you switched to a serial connection or Ethernet, your code would not agilent 34410a usb worked either because you were too specific on the interface. I should also point out that there is already a driver that exists for that instrument.
There must be something that I am not seeing. Computer Software Current Version: Contact an Expert Chat Live.
Communication with A via USB | Keysight Community
I tried many things but failed to get measurements from this device. I usv able to agilent 34410a usb with it fine via these two means. This program sets the A or A for 10 DCV measurements on the 10V range with a agilent 34410a usb second delay between the measurements.
Communication with 34410A via USB
They are two separate companies. See search results instead: BenchVue Software Platform BenchVue software offers multiple instrument measurement visibility and agilent 34410a usb capture from a PC or mobile device with no programming.
However, whenever I run my code, it gives me Err 7.
Hi Dennis Thanks for the reply. These commands are specific to a GPIB agilent 34410a usb only. I looked your code but its not needed to write a lot of lines. I looked through the manual and I couldn’t locate something specific for USB.